How confident are you about your cable connector cleanliness?
At the turn of the millennium, fiber inspection and determining cable connector cleanliness was performed with a microscope. This required technicians to use their eye on a potentially live fiber, risking personal injury every time they had to access fiber end face.
In mid-2005, the eye care community breathed a sigh of relief when the first fiber inspection probes made their way to the market. These probes were able to display an image of the endface on an LCD screen rather than directly on a tech’s retina. However, this image had to be interpreted. What constituted a defect or contaminant had to be based on user knowledge and gut instinct; depending on the quality of the focus, image centering, and several other parameters, there was always a chance of misinterpretation.
By 2010, the first intelligent analysis software for fiber inspection and cable connector cleanliness were created, which was based on the IEC standard. The software automatically detected and analyzed any defect, highlighted it on the display screen, and gave an overall pass-fail status, thereby removing the burden of interpretation and human error from the equation. Or did it?
Regardless of the power of the on-board intelligence, poor focus and poor image capture will lead to errors. More often than not, an out- of-focus speck, scratch, or trace will simply not appear on the screen. The intelligent software will give the connector a thumbs up, when in reality it should not have. This is what is referred to as a ‘false positive’. As the saying goes, garbage in, garbage out.
Figure 1 below compares manual centering and focusing to automatic centering and focusing. The methodology involved inspecting the connector, cleaning it, and inspecting it again (with the manual focus probe) until a pass was obtained. The same port was then inspected with an automated unit.